Problem Siemens 300 ventilator exp. Tidal vol. Values 3999 please tell
Background and Identification
The Siemens Servo 300 series of devices are a range of ventilators meant for adult, pediatric, and neonatal care. The devices offer increased patient comfort and treatment possibility through a mix or a Automode function as well as a range of ventilation modes.
The devices feature two main units connected via an interconnection cable, a control and patient unit. The control unit consists of several blocks meant to give the medical professional control over their patients care. Some controls on this unit include a patient range selection for tailoring the treatment for each specific use case, the respiratory pattern monitor, as well as Ventilation mode and Automode selection. The patient unit consists of the power unit as well as the inspiratory valve unit.
The control unit looks like several arrays of LED’s meant to show patient information, as well as several switches and knobs meant to control the device. The patient user uses transducers to measure flow and pressure.
It’s important to make sure your Siemens Servo 300 series device has the most up to date firmware and to know where to get spare parts to avoid common issues such as an exp valve failure.
Tidal Volume : 2 - 3999 ml
CMV Frequency : 5 – 150 BPM SIMV
Frequency : 1 – 40 BPM
Inspiratory Time : 10 -80 % of breathe cycle time
Pause Time : 0 – 30 % of breathe cycle time
Pressure Control : 0 – 100 CmH2O
Pressure Support : 0 – 100 CmH2O PEEP : 0 – 50 CmH2O
Trigger Sensitivity : 0 – (-) 17 in combinations with flow triggering.
Inspiratory Rise Time : 0 – 10% of breathe cycle time.
Gas Supply : Medical Grade Air & Oxygen 2 to 6.5 bar
Power Supply : 100, 120, 220, 240 VAC, 50/60 Hz Power Consumption : 50 W (standalone)
Battery : 2 built-in rechargeable 12 V, 1.9 Ah (External 24 VDC)
Back up time : 30 Min.
Source: Siemens Servo 300 Series Document
Siemens Servo 300 300A - Service manual.pdf